The arm joints on the television mounting system do not have a washer at the top joint which can result in excessive wear of the stop pins on the arm. This can allow the arm system and items connected to the arm to fall and injure the user or bystanders.
For additional information, contact TRUMPF Medical Systems Inc. toll-free at (888) 474-9359 between 8 a.m. and 5 p.m. ET Monday through Friday, or email Lindsey.email@example.com
The mounting systems were sold for use in hospital Intensive Care Units. The units are off-white and have multiple joints between the ceiling mount and the TV mount at the end of the arm system.
Secure the unit to prevent movement and contact TRUMPF Medical Systems for information on receiving a repair kit that includes the missing washers.
The firm received one report of a mounting system falling while being positioned. No injuries have been reported.
TRUMPF Medical Systems Inc. nationwide From January 2005 through December 2011 for about $2,100.
Ondal Medical Systems of America Inc., of Sandston, VA and TRUMPF Medical Systems Inc., of Charleston, SC
TRUMPF Medical Systems Inc., of Charleston, SC
The U.S. Consumer Product Safety Commission (CPSC) is charged with protecting the public from unreasonable risks of injury or death associated with the use of thousands of types of consumer products. Deaths, injuries, and property damage from consumer product incidents cost the nation more than $1 trillion annually. CPSC’s work to ensure the safety of consumer products has contributed to a decline in the rate of deaths and injuries associated with consumer products over the past 40 years.
Federal law bars any person from selling products subject to a publicly announced voluntary recall by a manufacturer or a mandatory recall ordered by the Commission.
For lifesaving information:
Please use the below phone number for all media requests.
Phone: (301) 504-7908
Spanish: (301) 504-7800