For additional information, contact TRUMPF Medical Systems Inc. toll-free at (888) 474-9359 between 8 a.m. and 5 p.m. ET Monday through Friday, or email Lindsey.email@example.com
Recall Details Report an Incident Involving this Product
The arm joints on the television mounting system do not have a washer at the top joint which can result in excessive wear of the stop pins on the arm. This can allow the arm system and items connected to the arm to fall and injure the user or bystanders.
The mounting systems were sold for use in hospital Intensive Care Units. The units are off-white and have multiple joints between the ceiling mount and the TV mount at the end of the arm system.
The firm received one report of a mounting system falling while being positioned. No injuries have been reported.
Secure the unit to prevent movement and contact TRUMPF Medical Systems for information on receiving a repair kit that includes the missing washers.
TRUMPF Medical Systems Inc. nationwide From January 2005 through December 2011 for about $2,100.
Ondal Medical Systems of America Inc., of Sandston, VA and TRUMPF Medical Systems Inc., of Charleston, SC
TRUMPF Medical Systems Inc., of Charleston, SC
The U.S. Consumer Product Safety Commission is charged with protecting the public from unreasonable risks of injury or death associated with the use of thousands of types of consumer products under the agency’s jurisdiction. Deaths, injuries, and property damage from consumer product incidents cost the nation more than $1 trillion annually. CPSC is committed to protecting consumers and families from products that pose a fire, electrical, chemical or mechanical hazard. CPSC's work to help ensure the safety of consumer products - such as toys, cribs, power tools, cigarette lighters and household chemicals -– contributed to a decline in the rate of deaths and injuries associated with consumer products over the past 40 years.
Federal law bars any person from selling products subject to a publicly-announced voluntary recall by a manufacturer or a mandatory recall ordered by the Commission.
To report a dangerous product or a product-related injury go online to www.SaferProducts.gov or call CPSC's Hotline at 800-638-2772 or teletypewriter at 301-595-7054 for the hearing impaired. Consumers can obtain news release and recall information at www.cpsc.gov, on Twitter @USCPSC or by subscribing to CPSC's free e-mail newsletters.
Please use the below phone number for all media requests.
Phone: (301) 504-7908
Spanish: (301) 504-7800